With the release of the DYNAMICS® software version 7.10, multimodal samples measured with a DynaPro® Plate Reader III or NanoStar® can be analyzed to quantify particle concentration. This adds an important tool for high-throughput screening of complex samples where different sized populations are present. For example, monomeric capsid concentration can be determined while simultaneously screening Adeno-associated virus (AAV) samples for aggregation during formulation and process development.
Figure 1 shows an example of the regularization graph in DYNAMICS, highlighting the particle concentration of each component in a mixture of two polystyrene latex beads with nominal radii of 10 and 50 nm.
To enable particle concentration in the DYNAMICS software, under the Parameters/Sample, set “Particle Concentration Calculation” to “Yes” and select the Particle Material and Shape from the list of options. The particle concentration of each peak is calculated according to the procedure detailed in a previous Ask an Expert article, “How do I use the new particle concentration feature in DYNAMICS?“ The calculation uses the hydrodynamic radius measured by dynamics light scattering (DLS), total static light scattering (SLS) intensity at the instrument’s scattering angle, and the Lorenz-Mie shape model for a solid sphere.
For monomodal samples it is assumed that all the particles in the scattering volume are the same size and shape in the particle concentration analysis, so the hydrodynamic radius Rh obtained by the cumulants fit is used. For multimodal samples, regularization analysis results are used and each peak is considered to be monodisperse, while the peak width indicates the uncertainty.
After running your samples, the particle concentration of each peak range appears under the regularization graph as shown in Figure 1. You can also setup the datalog grid to show particle concentrations for each peak range as shown in Figure 2.
The peak ranges DYNAMICS uses for the regularization analysis can be customized for multimodal samples with different size ranges as shown in Figure 3.
Limitations and important points:
To start using particle concentration calculations in your analyses, log on to the Support Center and download the latest version of the DYNAMICS software.
Do you have a question? Contact our experts here in Customer Support. We’re happy to help! Call +1 (805) 681-9009 option 4 or email email@example.com.